|R#||Site||Serial Number||Measurement||Start Date||Start Time||End Date||End Time||Data Quality Statement|
|1||BMS||Global Normal LI-200R||01/06/2017||00:00:00||01/17/2017||13:50:00||The global normal tracker has been having occasional tracking problems on various days during this time frame, data for the Global Normal CMP11 and LI-200R may be affected.
|2||BMS||Global Normal||01/06/2017||00:00:00||01/17/2017||13:50:00||The global normal tracker has been having occasional tracking problems on various days during this time frame, data for the Global Normal CMP11 and LI-200R may be affected.
|3||BMS||Global Normal LI-200R||12/01/2016||13:00:00||12/05/2016||12:26:00||When the LI-200R was first installed the data logger output was accidentally set to sample instead of average during this time, so data was 1-minute samples instead of 1-minute averages, as a result data will disagree with other instruments during cloud passage.|
|4||BMS||Global Normal LI-200R||12/01/2016||13:00:00||12/14/2016||08:44:00||When the LI-200R was first installed the calibration factor at 45 degrees was used instead of the preferred 0 degrees for global normal:CF@45=91.575 W/mē/mV; CF@0=91.707 W/mē/mV.
|5||BMS||140840||Global Normal||07/04/2016||00:00:00||07/05/2016||14:00:00||The LICOR LI-2020 tracker is not working during this time, data should not be used during this time.|
|6||BMS||140840||Global Normal||01/17/2016||00:00:00||01/18/2016||15:30:00||The LICOR LI-2020 tracker is not working during this time, data should not be used during this time.|
|7||BMS||140840||Global Normal||09/21/2015||00:00:00||09/21/2015||14:20:00||The LICOR LI-2020 tracker is not working during this time, instrument is facing northward/downward.|
|8||BMS||140840||Global Normal||08/12/2015||12:00:00||08/28/2015||14:00:00||The LICOR LI-2020 tracker for the Global Normal CMP11 has failed. Instrument is pointed in a somewhat southeasterly direction. Tracker was replaced on 8/28/2015.|
|9||BMS||920056||Global Normal||09/12/2014||00:00:00||09/15/2014||09:15:00||The CM21 was found on 9/12/2014 with moisture inside the inner and outer dome. Instrument was removed from service on 9/15/2014 at 08:53 MST and replaced with the calibration spare (it was due to be swapped out anyway).
|10||BMS||Global Normal||05/08/2013||08:29:00||05/09/2013||12:59:00||The global normal signal cable was being replaced during this time. Data is bogus.|
|11||BMS||Global Normal||05/06/2013||18:30:00||05/06/2013||21:30:00||The global normal signal cable is breaking and reporting bogus data during this time.|
|12||BMS||910056||Global Normal||08/16/2011||00:00:00||08/16/2011||12:10:00||Solar tracker was not tracking the sun and was fixed in an easterly direction. Tracker was reset at approx. 12:10 MST.|
|13||BMS||910002||Global Normal||03/06/2011||10:43:00||03/09/2011||12:11:00||Signal is negative or bogus during this time. Bendix connector at instrument was re-seated which seemed to fix the problem.|
|14||BMS||Global Normal||09/05/2010||00:00:00||09/07/2010||09:11:00||The LICOR tracker for the Global Normal CM-21 was not functioning during this time. Data is bad.|
|15||BMS||Global Normal||12/16/2002||14:43:00||12/17/2002||10:57:00||The J-Box on the BMS Radiometer Tower was being replaced, data from this instrument is missing or bogus.|
|16||BMS||Global Normal||11/01/2000||00:00:00||11/07/2000||The LI-COR LI-2020 Solar Tracker is powered by a PV panel with backup battery, however the charging cycle does not work correctly. Therefore the battery does not recharge after cloud cover and the tracker stops. The battery has to be (manually) swapped with another charged battery, until the problem is corrected.|