Total Records:

R# | Serial Number | Calibration Facility | Calibration Date | Date for next Calibration | Responsivity [45] | Calibration Factor [45] | Instrument::Make | Instrument::Model | Instrument::Description | Instrument::Program | Instrument::Factory Calibration | Instrument::Date of Factory Calibration | Responsivity Bins [2 deg] | Responsivity Function Coefficients | Other Calibrations | Calibration File | Contact Person | Documentation Title | Calibration Method | Calibration Notes | Other Information | Datalogger Entry Date | Responsivity [45] [Corr] | Calibration Factor [45] [Corr] | Correction Method | Responsivity Bins Average [2 deg] |
---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|

1 | A360 | Delta-T | 05/18/2009 | 05/18/2011 | Delta-T | SPN1 | Pyranometer | NREL/Program | Cf=1 W/m²/mV | 05/18/2009 | 08/02/2010 | |||||||||||||||

2 | A360 | SRRL/2011-02 | 05/05/2011 | 05/05/2012 | R@45=991.09 µV/W/m² ; +9.53/-3.79% ; N=11 ; Z:30-60 | CF@45=1.0090 W/m²/mV | Delta-T | SPN1 | Pyranometer | NREL/Program | Cf=1 W/m²/mV | 05/18/2009 | [A360_Delta-T_SPN1.2deg.txt] | [A360_Delta-T_SPN1.coef.txt] | [A360_Delta-T_SPN1.lato.txt] | Tom Stoffel | BORCAL 2011-02 Certificates | For BMS, Diffuse CF was entered as 1.0 | 05/26/2011 | R@45=991.09 µV/W/m² ; +9.53/-3.79% ; N=11 ; Z:30-60 | CF@45=1.0090 W/m²/mV | None ; Rnet=0.0 | ||||

3 | A360 | SRRL/2012-01 | 05/04/2012 | 05/04/2013 | R@45=994.26 µV/W/m² ; +8.99/-4.38% ; N=12 ; Z:30-60 | CF@45=1.0058 W/m²/mV | Delta-T | SPN1 | Pyranometer | NREL/Program | Cf=1 W/m²/mV | 05/18/2009 | [A360_Delta-T_SPN1.2deg.txt] | [A360_Delta-T_SPN1.coef.txt] | [A360_Delta-T_SPN1.lato.txt] | Tom Stoffel | BORCAL 2012-01 Certificates | For BMS, CF was used for both Global and Diffuse | 06/07/2012 | R@45=994.26 µV/W/m² ; +8.99/-4.38% ; N=12 ; Z:30-60 | CF@45=1.0058 W/m²/mV | None ; Rnet=0.0 | ||||

4 | A360 | SRRL/2013-02 | 06/20/2013 | 06/20/2014 | R@45=1008.7 µV/W/m² ; +5.56/-4.89% ; N=8 ; Z:30-60 | CF@45=0.99134 W/m²/mV | Delta-T | SPN1 | Pyranometer | NREL/Program | Cf=1 W/m²/mV | 05/18/2009 | [A360_Delta-T_SPN1.2deg.txt] | [A360_Delta-T_SPN1.coef.txt] | [A360_Delta-T_SPN1.lato.txt] | Tom Stoffel | BORCAL 2013-02 Certificates | 07/17/2013 | R@45=1008.7 µV/W/m² ; +5.56/-4.89% ; N=8 ; Z:30-60 | CF@45=0.99134 W/m²/mV | None ; Rnet=0.0 | |||||

5 | A360 | SRRL/2014-01 | 05/19/2014 | 05/19/2015 | R@45=1009.5 µV/W/m² ; +6.66/-6.13% ; N=6 ; Z:30-60 | CF@45=0.99057 W/m²/mV | Delta-T | SPN1 | Pyranometer | NREL/Program | Cf=1 W/m²/mV | 05/18/2009 | [A360_Delta-T_SPN1.2deg.txt] | [A360_Delta-T_SPN1.coef.txt] | [A360_Delta-T_SPN1.lato.txt] | Afshin M. Andreas | BORCAL 2014-01 Certificates | 06/04/2014 | R@45=1009.5 µV/W/m² ; +6.66/-6.13% ; N=6 ; Z:30-60 | CF@45=0.99057 W/m²/mV | None ; Rnet=0.0 | [A360_Delta-T_SPN1.2avg.txt] |