Total Records:

R# | Serial Number | Calibration Facility | Calibration Date | Date for next Calibration | Responsivity [45] | Calibration Factor [45] | Instrument::Make | Instrument::Model | Instrument::Description | Instrument::Program | Instrument::Factory Calibration | Instrument::Date of Factory Calibration | Responsivity Bins [2 deg] | Responsivity Function Coefficients | Other Calibrations | Calibration File | Contact Person | Documentation Title | Calibration Method | Calibration Notes | Other Information | Datalogger Entry Date | Responsivity [45] [Corr] | Calibration Factor [45] [Corr] | Correction Method | Responsivity Bins Average [2 deg] |
---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|---|

1 | 18041F3 | Eppley | 7.68 | Eppley | PSP | Precision Spectral Pyranometer | NREL/Program | 7.68 tg | Tom Kirk | |||||||||||||||||

2 | 18041F3 | NREL/98-05 | 09/03/1998 | 09/03/1999 | Rs=6.793 uV/W/sqm | Eppley | PSP | Precision Spectral Pyranometer | NREL/Program | 7.68 tg | Tom Stoffel | Borcal98-05 | Cf=147.21 W/sqm/mV =/-2.2% | |||||||||||||

3 | 18041F3 | SRRL/2003-01 | 07/02/2003 | 07/02/2004 | RS@45=6.6774 µV/W/m² ; +3.19/-4.55% ; N=8 | CF@45=149.76 W/m²/mV | Eppley | PSP | Precision Spectral Pyranometer | NREL/Program | 7.68 tg | [18041F3_Eppley_PSP.2deg.txt] | [18041F3_Eppley_PSP.coef.txt] | Tom Stoffel | Borcal 2003-01 | Valid Incident angle range for RS@45: 30-60 | 08/28/2003 | |||||||||

4 | 18041F3 | SRRL/2005-01 | 05/19/2005 | 05/19/2006 | RS@45=6.6263 µV/W/m² ; +3.19/-4.38% ; N=10 | CF@45=150.91 W/m²/mV | Eppley | PSP | Precision Spectral Pyranometer | NREL/Program | 7.68 tg | [18041F3_Eppley_PSP.2deg.txt] | [18041F3_Eppley_PSP.coef.txt] | Tom Stoffel | Borcal 2005-01 | Valid Incident angle range for RS@45: 30-60 | 06/02/2005 | |||||||||

5 | 18041F3 | SRRL/2007-01 | 06/08/2007 | 06/08/2008 | RS@45=6.6421 µV/W/m² ; +2.90/-5.02% ; N=13 ; I:30-60 | CF@45=150.55 W/m²/mV | Eppley | PSP | Precision Spectral Pyranometer | NREL/Program | 7.68 tg | [18041F3_Eppley_PSP.2deg.txt] | [18041F3_Eppley_PSP.coef.txt] | [18041F3_Eppley_PSP.lato.txt] | Tom Stoffel | BORCAL 2007-01 Certificates | 06/28/2007 | n/a ; n/a ; n/a ; I:n/a | n/a | None ; Estimated thermal offset error = -20 W/m² | ||||||

6 | 18041F3 | SRRL/2008-04 | 08/18/2008 | 08/18/2009 | RS@45=6.5770 µV/W/m² ; +3.18/-6.35% ; N=13 ; I:30-60 | CF@45=152.05 W/m²/mV | Eppley | PSP | Precision Spectral Pyranometer | NREL/Program | 7.68 tg | [18041F3_Eppley_PSP.2deg.txt] | [18041F3_Eppley_PSP.coef.txt] | [18041F3_Eppley_PSP.lato.txt] | Tom Stoffel | BORCAL 2008-04 Certificates | n/a ; n/a ; n/a ; I:n/a | n/a | None ; Estimated thermal offset error = -20 W/m² | |||||||

7 | 18041F3 | SRRL/2009-01 | 05/27/2009 | 05/27/2010 | RS@45=6.5067 µV/W/m² ; +3.80/-4.49% ; N=12 ; I:30-60 | CF@45=153.69 W/m²/mV | Eppley | PSP | Precision Spectral Pyranometer | NREL/Program | 7.68 tg | [18041F3_Eppley_PSP.2deg.txt] | [18041F3_Eppley_PSP.coef.txt] | [18041F3_Eppley_PSP.lato.txt] | Tom Stoffel | BORCAL 2009-01 Certificates | n/a ; n/a ; n/a ; I:n/a | n/a | None ; Estimated thermal offset error = -20 W/m² | |||||||

8 | 18041F3 | SRRL/2011-02 | 05/05/2011 | 05/05/2012 | R@45=6.6411 µV/W/m² ; +3.11/-4.46% ; N=14 ; Z:30-60 | CF@45=150.58 W/m²/mV | Eppley | PSP | Precision Spectral Pyranometer | NREL/Program | 7.68 tg | [18041F3_Eppley_PSP.2deg.txt] | [18041F3_Eppley_PSP.coef.txt] | [18041F3_Eppley_PSP.lato.txt] | Tom Stoffel | BORCAL 2011-02 Certificates | 02/21/2013 | R@45=6.6411 µV/W/m² ; +3.11/-4.46% ; N=14 ; Z:30-60 | CF@45=150.58 W/m²/mV | Effective Net IR ; Rnet=0.600 µV/W/m² ; Estimated |